- Case report
- Open Access
Delayed tracheal extubation after cardiac surgery due to cardiogenic ventilator auto-triggering: a case report
JA Clinical Reports volume 7, Article number: 55 (2021)
Ventilator auto-triggering is associated with poor outcomes. Herein, we present a case of delayed tracheal extubation after cardiac surgery due to cardiogenic auto-triggering.
A 73-year-old male with chronic constrictive pericarditis underwent radical pericardiectomy. After confirming hemodynamic stability, we conducted spontaneous breathing trial (SBT) with a flow-trigger sensitivity of 1 L/min. As his respiratory rate (RR) increased to more than 60 breaths/min and tidal volume decreased to less than 100 mL, this SBT was considered a failure. Next morning, SBT was reperformed and the result was unchanged. However, we noticed that his heart rate and RR were the same and suspected auto-triggering caused by cardiogenic oscillations. We changed ventilator mode from flow triggering to pressure triggering of −2 cmH2O and he was uneventfully extubated.
We experienced a case of delayed tracheal extubation after cardiac surgery due to cardiogenic auto-triggering. Auto-triggering can be reduced by changing ventilator trigger mode.
Ventilator auto-triggering, a type of patient-ventilator dyssynchrony, is an inappropriate triggering of a ventilator breath in the absence of spontaneous inspiratory effort. This phenomenon can be caused by circuit leaks, water condensation in the circuit, or cardiogenic oscillations . Herein, we present a case of delayed tracheal extubation after cardiac surgery due to ventilator auto-triggering caused by cardiogenic oscillations.
The patient was a 73-year-old male (height, 169 cm; weight, 71 kg) with chronic constrictive pericarditis. He underwent radical pericardiectomy and was postoperatively transferred to the intensive care unit (ICU) under propofol sedation with tracheal intubation. Postoperative observation was conducted under continuous intravenous infusion of propofol (1 mg/kg/h), fentanyl (20 μg/h), and dexmedetomidine (0.4 μg/kg/h) while maintaining hemodynamic stability. His ventilator settings on synchronized intermittent mandatory ventilation (pressure control) [SIMV(PC)] + pressure support ventilation (PSV) were as follows: a driving pressure of 15 cmH2O, support pressure of 15 cmH2O, respiratory rate (RR) of 12 breaths/min, fraction of inspiratory oxygen (FIO2) of 40%, positive end-expiratory pressure (PEEP) of 5 cmH2O, and flow-trigger sensitivity of 1 L/min. After confirming hemodynamic stability, we decreased driving pressure and support pressure to 10 cmH2O before starting a spontaneous breathing trial (SBT). Then, his spontaneous breaths were triggered 5 breaths/min and tidal volume was more than 700 mL when spontaneous breaths were triggered. Although RR was low due to relative deep sedation, the patient’s other status fulfilled the criteria for starting a SBT. In addition, we thought that if a SBT was started, his PaCO2 became higher and his RR would increase. We conducted a SBT with a PSV of 5 cmH2O, PEEP of 5 cmH2O, FIO2 of 40%, and a flow-trigger sensitivity of 1 L/min, and discontinued the continuous intravenous infusion of propofol. Subsequently, his RR increased to more than 60 breaths/min and tidal volume decreased to less than 100 mL (Fig. 1); this SBT was considered a failure and the prior SIMV(PC) +PSV ventilator settings were restored. The next morning, SBT was reperformed, and the result was the same as the day before. This time, however, we noticed that his heart rate (HR) and RR were exactly the same (Fig. 2), at 60/min, and suspected ventilator auto-triggering caused by cardiogenic oscillations. To examine this hypothesis, we decreased flow-trigger sensitivity to 2 L/min during, but the patient’s respiratory pattern was unchanged. Next, we changed the ventilator mode from flow triggering of 1 L/min to pressure triggering of −2 cmH2O. After that, his RR became 6 breaths/min and his tidal volume greater than 600 mL (Fig. 3). Upon discontinuation of the anesthetic drugs, the patient promptly emerged from anesthesia and was uneventfully extubated. Following extubation, his RR was about 12 breaths/min and he could breathe deeply. The subsequent postoperative course was uneventful, and he was discharged on postoperative day 15. We obtained a written informed consent from the patient for publication of this case report.
We present a case of delayed tracheal extubation after cardiac surgery due to ventilator auto-triggering caused by cardiogenic oscillations. We did not recognize the auto-triggering during the first SBT because there were a few spontaneous inspiratory efforts due to relative deep sedation; hence, the cardiogenic oscillation-triggered breaths simply looked like shallow, fast breathing. However, during the second SBT, we were able to notice that his HR and RR were the same by carefully observing his vital signs; at that point, we suspected ventilator auto-triggering caused by cardiogenic oscillations. We could definitively diagnose auto-triggering by changing the trigger setting from flow-trigger to pressure-trigger. Clinicians should consider the possibility of auto-triggering caused by cardiogenic oscillations when performing a SBT before the appearance of sufficient spontaneous breathing.
Patient-ventilator dyssynchrony including auto-triggering, which is caused by circuit leaks, water condensation in the circuit, or cardiogenic oscillations, is reported to occur in 26-82% patients . Cardiogenic auto-triggering tends to occur in patients with brain death and those who have just undergone cardiac surgery [3, 4]. Patients with brain death tends to have hyperdynamic cardiovascular state , which may cause cardiogenic ventilator auto-triggering. Indeed, auto-triggering is reported to occur more often in patients with hyperdynamic cardiovascular state after cardiac surgery . If not detected, this phenomenon can cause prolonged duration of mechanical ventilation, prolonged ICU and hospital stays, and higher ICU and hospital mortality [6,7,8]. Thus, clinicians must be aware of this possibility, particularly in critically ill patients.
The mechanism of cardiogenic oscillation is not completely clear, but there are several possible contributory factors. Pulmonary artery pulsatility is reported to be the main cause of cardiogenic oscillations . Additionally, changes in heart volume during systole and diastole may change intrathoracic pressure, which may change airway pressure or cause compression and expansion of the lung . Thus, enlargement of the heart may be associated with cardiogenic oscillation. In the present case, this patient’s postoperative cardiothoracic ratio (CTR) was 67%, indicating enlargement of heart. Indeed, cardiogenic oscillations are reported to occur in patients with larger CTR values .
Our PubMed search could not reveal any reports describing a relationship between cardiac oscillation and chronic constrictive pericarditis or radical pericardiectomy. However, hyperdynamic cardiovascular state after radical pericardiectomy may be associated with cardiac oscillation, because, as mentioned above, cardiac oscillation tends to occur in patients with hyperdynamic cardiovascular state . Indeed, this patient’s cardiac output after surgery was more than 4.2 L/min.
One way to terminate false triggering is to change ventilator settings from the very sensitive “flow trigger” mechanism to the less sensitive “pressure trigger” mechanism . In the present case, we could definitively diagnose auto-triggering by changing from the flow-trigger to the pressure-trigger setting. On the other hand, clinicians have to be careful for mistriggering, which can cause prolonged duration of mechanical ventilation, when using pressure trigger. Thus, clinicians have to select which triggering setting to use according to the patient’s situation.
We experienced a case of delayed tracheal extubation after cardiac surgery due to cardiogenic ventilator auto-triggering. Auto-triggering can be reduced by decreasing ventilator trigger sensitivity. As patient-ventilator dyssynchrony may affect patients’ outcomes, early recognition is extremely important for clinicians. Clinicians have to be careful of auto-triggering when performing a SBT before the appearance of sufficient spontaneous breathing.
Availability of data and materials
Intensive care unit
Synchronized intermittent mandatory ventilation
Pressure support ventilation
- FIO2 :
Fraction of inspiratory oxygen
Positive end-expiratory pressure
Spontaneous breathing trial
Tobin MJ, Jubran A, Laghi F. Patient-ventilator interaction. Am J Respir Crit Care Med. 2001;163(5):1059–63. https://doi.org/10.1164/ajrccm.163.5.2005125.
Gilstrap D, MacIntyre N. Patient-ventilator interactions. Implications for clinical management. Am J Respir Crit Care Med. 2013;188(9):1058–68. https://doi.org/10.1164/rccm.201212-2214CI.
Arbour R. Cardiogenic oscillation and ventilator autotriggering in brain-dead patients: a case series. Am J Crit Care. 2009;18(496):488–95.
Imanaka H, Nishimura M, Takeuchi M, Kimball WR, Yahagi N, Kumon K. Autotriggering caused by cardiogenic oscillation during flow-triggered mechanical ventilation. Crit Care Med. 2000;28(2):402–7. https://doi.org/10.1097/00003246-200002000-00019.
Arbour R. Confounding factors in brain death: cardiogenic ventilator autotriggering and implications for organ transplantation. Intensive Crit Care Nurs. 2012;28(6):321–8. https://doi.org/10.1016/j.iccn.2012.03.003.
Chao DC, Scheinhorn DJ, Stearn-Hassenpflug M. Patient ventilator trigger asynchrony in prolonged mechanical ventilation. Chest. 1997;112(6):1592–9. https://doi.org/10.1378/chest.112.6.1592.
de Wit M, Miller KB, Green DA, Ostman HE, Gennings C, Epstein SK. Ineffective triggering predicts increased duration of mechanical ventilation. Crit Care Med. 2009;37(10):2740–5. https://doi.org/10.1097/ccm.0b013e3181a98a05.
Blanch L, Villagra A, Sales B, Montanya J, Lucangelo U, Luján M, et al. Asynchronies during mechanical ventilation are associated with mortality. Intensive Care Med. 2015;41(4):633–41. https://doi.org/10.1007/s00134-015-3692-6.
Suarez-Sipmann F, Santos A, Peces-Barba G, Bohm SH, Gracia JL, Calderón P, et al. Pulmonary artery pulsatility is the main cause of cardiogenic oscillations. J Clin Monit Comput. 2013 Feb;27(1):47–53. https://doi.org/10.1007/s10877-012-9391-8.
West JB, Hugh-Jones P. Pulsatile gas flow in bronchi caused by the heart beat. J Appl Physiol. 1961;16(4):697–702. https://doi.org/10.1152/jappl.19220.127.116.117.
Wijdicks EF, Manno EM, Holets SR. Ventilator self-cycling may falsely suggest patient effort during brain death determination. Neurology. 2005;65(5):774. https://doi.org/10.1212/01.wnl.0000174626.94197.62.
The authors thank Yoshikazu Nikaido, PhD (Department of Anesthesiology, Hirosaki University Graduate School of Medicine, Aomori, Japan) for his valuable comments.
Ethics approval and consent to participate
Consent for publication
Informed consent for scientific publication was obtained from the patients.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Takekawa, D., Uchida, S. & Hirota, K. Delayed tracheal extubation after cardiac surgery due to cardiogenic ventilator auto-triggering: a case report. JA Clin Rep 7, 55 (2021). https://doi.org/10.1186/s40981-021-00458-4
- Ventilator auto-triggering
- Cardiogenic oscillation
- Delayed tracheal extubation