- Clinical research letter
- Open Access
Comparison of inflationary non-invasive blood pressure (iNIBP) monitoring technology and conventional deflationary non-invasive blood pressure (dNIBP) measurement in detecting hypotension during cesarean section
© The Author(s) 2018
- Received: 6 September 2017
- Accepted: 27 December 2017
- Published: 5 January 2018
The Nihon Kohden linear inflationary non-invasive blood pressure (iNIBP) monitoring technology is an oscillometric device that measures blood pressure by detecting oscillations during inflation. Systolic blood pressure can be recorded without overinflating the cuff higher than the true systolic pressure. Thus, total time taken for inflation and deflation is shorter than that by the conventional deflation devices. In this study, the ability of iNIBP to detect maternal hypotension during cesarean section faster than deflationary non-invasive blood pressure (dNIBP) monitoring devices under clinical settings was evaluated prospectively.
A prospective study of singleton planned cesarean sections at a tertiary center was conducted from August 2015 to April 2016. The combined spinal-epidural anesthesia (CSEA) technique through a single puncture was performed for cesarean section at the center where the study was carried out. An iNIBP cuff was placed on the same arm as the intravenous line, and a dNIBP cuff was placed on the other arm. Due to left uterine displacement by approximately 10° tilt of OR table, hypotension in this study was defined as systolic pressure of 107 mmHg or less, when measured in the left arm, which was about 10 cm lower, and pressure of 92 mmHg or less in the right arm which was about 10 cm higher. This setup was done to evaluate which device detected hypotension faster under clinical settings. A two-tailed Z test was performed to statistically analyze the difference between iNIBP and dNIBP measurement results.
One hundred singleton planned cesarean deliveries under CSEA were included after 36 weeks of gestation. Out of the 100, 76 women (76%) experienced maternal hypotension. Of these, iNIBP detected hypotension faster than dNIBP in 47 cases (61.8%).
It was found that iNIBP detected hypotension faster than conventional dNIBP without compromising the reliability of measurement. This may lead to early treatment of maternal hypotension and prevention of adverse events related to the mother and the fetus.
- Cesarean section
- Maternal hypotension
- Combined spinal-epidural anesthesia
Nowadays, spinal anesthesia is the first choice for anesthesia during cesarean sections. Pregnant women tend to experience a drop in blood pressure because of supine hypotensive syndrome and decreased peripheral vascular resistance. Maternal hypotension is an adverse event for the mother and the fetus and can cause fetal bradycardia . Therefore, it is important to detect maternal hypotension as soon as possible and control it promptly. However, it often takes time to measure blood pressure because of noise and change in body position in clinical practice.
The Nihon Kohden linear inflationary non-invasive blood pressure (iNIBP®) measurement technology is an oscillometric device used to measure blood pressure by detecting oscillations during inflation. We can read systolic blood pressures without the need to overinflate the cuff higher than the true systolic pressure. Thus, the total time taken for inflation and deflation is shorter than that taken by conventional deflationary devices.
In this study, we prospectively evaluated whether iNIBP can detect maternal hypotension during spinal anesthesia earlier than deflationary non-invasive blood pressure (dNIBP) in clinical settings in which noise and vibration could lead to inaccurate measurement, requiring re-measurement of blood pressure.
Approval of the Institutional Review Broad and written informed consent from the patients were obtained for this study. We conducted a prospective study of singleton planned cesarean sections at a tertiary center from August 2015 to April 2016 (UMIN000027385).
At our center, the combined spinal-epidural anesthesia (CSEA) technique is performed for cesarean sections. The L3-4 interspace was selected. The CSEA was initiated with 0.5% hyperbaric bupivacaine (8 mg) and fentanyl (20 μg) via a 27-gauge spinal needle placed via a 16-gauge Tuohy needle, and the epidural catheter was inserted.
A two-sided Z test was performed to statistically analyze the differences between the iNIBP and dNIBP measurements.
We included 100 singleton planned cesarean deliveries under spinal CSEA, after 36 weeks of gestation.
Maternal demographic and clinical characteristics
Mean ± SD or median (range)
34.7 ± 5.0
GA at delivery (weeks)
24.7 ± 2.8
Comparison of the speeds of detection of hypotension by different devices
The number of cases in which hypotension was detected faster (n (%))
We conducted the prospective study to evaluate which method between iNIBP and dNIBP could detect the post-spinal maternal hypotension during cesarean section faster. It was found that iNIBP could detect maternal hypotension faster even under clinical settings, in which noise and vibration could lead to inaccurate measurement and require re-measurement of blood pressure.
The spinal anesthesia for cesarean section is likely to cause maternal hypotension because of the spine hypotensive syndrome and decreased peripheral vascular resistance. Hypotension during cesarean section is defined as systolic blood pressure ≤ 100 mmHg or ≤ 20% below the baseline blood pressure . The sustained maternal hypotension affects not only maternal circulation but also uteroplacental blood flow. As it may cause fetal bradycardia and acidosis , it needs to be prevented and treated promptly. Left uterine displacement and fluid management by colloid co-loading are the preventive measures employed .
The conventional non-invasive blood pressure monitoring devices are designed to detect oscillations during deflation. Therefore, they require elevation of the cuff pressure higher than the true systolic pressure. This excessive cuff pressure may cause adverse events such as pain, venous stasis, thrombophlebitis, peripheral neuropathy, and compartment syndrome [4, 5]. Meanwhile, the iNIBP is designed to detect oscillations slowly during inflation and reduce the cuff pressure at the time of detecting the systolic blood pressure. Thus, it requires lower cuff pressure and shorter time to detect blood pressure as compared to dNIBP .
Furthermore, previous studies suggested that the hemodynamic changes caused by pregnancy and pre-eclampsia may attribute to inaccuracy of the devices. Immediate detection of the oscillations during inflation may provide accurate measurements without transmission of the pressure wave. Therefore, inflationary oscillometry may be more accurate than deflationary oscillometry and mercury sphygmomanometry in pregnancy and pre-eclampsia cases [7, 8].
A limitation of this study is its small sample size. Further research is required to have a clear understanding of the advantages of iNIBP for cesarean section cases. A further limitation was that blood pressure was measured not every minute but 2 min. The BP measurement every minute until delivery of the baby is recommended.
Through this study, we conclude that iNIBP detects hypotension faster than conventional dNIBP, without compromising the reliability of measurement. This may lead to early treatment of maternal hypotension and prevention of maternal and fetal adverse events.
This study was not funded.
Both authors read and approved the final manuscript.
Ethics approval and consent to participate
Approval of the Institutional Review Broad was obtained.
Consent for publication
The written informed consent from the patients was obtained for this study.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Ebner H, Barcohana J, Bartoshuk AK. Influence of postspinal hypotension on the fetal electrocardiogram. Am J Obstet Gynecol. 1960;80:569–72.View ArticlePubMedGoogle Scholar
- Corke BC, Datta S, Ostheimer GW, Weiss JB, Alper MH. Spinal anaesthesia for caesarean section. The influence of hypotension on neonatal outcome. Anaesthesia. 1982;37:658–62.View ArticlePubMedGoogle Scholar
- American Society of Anesthesiologists Task Force on Obstetric Anesthesia: Practice guidlines for obstetric anesthesia: an upload report by the American Society of Anesthesiologists Task Force on Obstetirc Anesthesia. Anesthesiology. 2016:10.Google Scholar
- Bickler PE, Schapera A, Bainton CR. Acute radial nerve injury from use of an automatic blood pressure monitor. Anesthesiology. 1990;73:186–8.View ArticlePubMedGoogle Scholar
- Sutin KM, Longaker MT, Wahlander S, Kasabian AK, Capan LM. Acute biceps compartment syndrome associated with the use of a noninvasive blood pressure monitor. Anesth Analg. 1996;83:1345–6.View ArticlePubMedGoogle Scholar
- Onodera J, Kotake Y, Fukuda M, et al. Validation of inflationary non-invasive blood pressure monitoring in adult surgical patients. J Anesth. 2011;25:127–30.View ArticlePubMedGoogle Scholar
- Golara M, Benedict A, Jones C, Randhawa M, Poston L, Shennan AH. Inflationary oscillometry provides accurate measurement of blood pressure in pre-eclampsia. BJOG. 2002;109:1143–7.View ArticlePubMedGoogle Scholar
- de Greeff A, Beg Z, Gangji Z, Dorney E, Shennan AH. Accuracy of inflationary versus deflationary oscillometry in pregnancy and preeclampsia: OMRON-MIT versus OMRON-M7. Blood Press Monit. 2009;14:37–40.View ArticlePubMedGoogle Scholar